HP 6632B PDF
The HP B is a single channel W DC programmable power supply with GPIB and RS connectivity. It is part of the HP x series. Agilent B/B/B/B System DC Power. Supplies All references to HP-IB have been changed to GPIB. A note about. I just picked up two HP B power supplies as I read somewhere quickly that it not only can be programmed via GPIB to set voltage/current.
|Published (Last):||15 September 2010|
|PDF File Size:||2.69 Mb|
|ePub File Size:||13.49 Mb|
|Price:||Free* [*Free Regsitration Required]|
Unless otherwise noted, specifications apply when measured atthe rear terminals after a minute warm-up period. The calibration procedure can only be performed using the SCPI languagecommands. Page 44 5 – Front Panel OperationAs previously mentioned, the HP A dc source has the capability of measuring outputwaveform parameters such as peak, minimum, high level, and low level as illustrated in thefollowing figure.
CError MessagesError Number ListThis appendix gives the error numbers and descriptions that are returned by the dc source. One 6632bb thing is that the current meter on it is fairly precise. If you read through the manual, you know, that it can also behave as current sink load. Theoutput dropped to zero, CV turned off, and Protturned on.
Page 22 3 – InstallationLocationThe outline diagram in figure gives the dimensions of your dc source.
Providing the finest test equipment solutions since 1992
Readings within high currentlimits see tables B-3,4, 5 or 6. After much swearing, compressed air, and windex, the inside 6632h the chassis looked like this:.
East Dane Designer Men’s Fashion. Alexa Actionable Analytics for the Web.
In this example, OC indicates thatan overcurrent condition has occurred. Page 75 Manual UpdatesThe following updates have been made to this manual since the November printingindicated on the Printing History page. Using the online preview, you can quickly view the contents and go to the page where you will find the solution to your problem with HP Hewlett-Packard B.
Agilent / HP B Watt DC System Power Supply 20v 5a | eBay
Your manual failed to upload Shock Hazard Operating personnel must not remove instrument covers. US and upHP B: System KeysThis is the blue, unlabeled key, which is also shown 66332b Shift in this guide. CC The dc source output is in constant-current mode. Electronics AdviceTest Equipment.
One of the cool things about these supplies is that they talk RS in many languages. In this mode the portfunctions as a ph inhibit input with a discrete fault indicator output signal. B – Verification and CalibrationFront Panel CalibrationThese procedures assume you understand how to operate front panel keys see chapter 5.
Please enter a question. I just got it from a good friend of mine, who owned two of these fine instruments. I have two such supplies and they are really cool. See All Buying Options. Sorry, manual briefs data are unavailable at this moment. Using the Keypad shift pandSome of the front panel keys perform two functions, one labeled in black and theother in blue.
HP (Hewlett-Packard) B user manual – – Solve your problem
Amazon Restaurants Food delivery from local restaurants. Page 24 3 – InstallationVoltage DropsThe load wires must also be large enough to avoid excessive voltage drops due to the impedance ofthe wires.
Page 56 B – Verification and Calibration5. Enter the calibration password from Entry keypad and press Enter. I used a similar Philips Power supply source before PEbut this thing is stuffed with only imprecise analog meters and is not comparable with this fine HP voltage supply, and one does have to use an additional precise voltage meter I do use a Fluke A Multimeter, a quite old, but freshly calibrated, still precise voltmeter.
The verification procedures do not check all the operating parameters, but verify that the dc sourceis performing properly. Press Shift, OV,Enter8. In general, if the wires are heavy enough to carry the maximum short circuit currentwithout overheating, excessive voltage drops will not be a problem.
Measurement samples may notcoincide with the actual maximumor minimum point of the waveform. You do not have to do a complete calibration each time. Set the instrument to the operating state that you want to save. ComiXology Thousands of Digital Comics.
TICS International Ltd., UK
B – Verification and Calibration5. On the Function keypad, press Protect. Read the application-note about testing the solar-cells. Page 32 4 – Turn-On CheckoutCheckout ProcedureThe test in this section checks for output voltage and current on the dc source.
Amazon Rapids Fun stories for kids on the go.