BGI 5012 PDF
BGI frmOMNI™ Ambient Air Sampler (Filter Reference FRM Omni (Sampler Only) and Power Supply. Accessories / Spare Parts. BGI BGI Internal Errors. BAO Process Name. BGI Unable to create .:VIP_BGI_Core_Incident:Inbound:incCreateTask. BGI husec41_c FIG CDS – 0 ID=fig|peg;Name= Thiamine biosynthesis protein thiI. husec41_c FIG CDS
|Published (Last):||9 October 2015|
|PDF File Size:||9.91 Mb|
|ePub File Size:||1.86 Mb|
|Price:||Free* [*Free Regsitration Required]|
Record in check sheet.
Hitting the Mark | Ignite Sport
Retrieved from ” http: It is desirable to run for at least 3 hours with filtered air to assess the noise level. For the zero check attach a Hepa cartridge filter to the inlet, scroll down to the Service menu, enable the keys and proceed to open the head and advance the filter, stop the motor and close the head.
BGI works alongside various youth workers and tutors to provide opportunities for young people in areas such as:.
With a good core group of them choosing to come along regularly. Return to Standard Operating Procedures.
Scroll back to normal operation. Views Page Discussion View source History. Operational flow rate is bg to This approach allows use of aerosol radiative transfer theory to essentially remove scattering effects that otherwise impact on the derived light absorption coefficient; the data inversion algorithm takes multiple scattering processes inside the deposited aerosol and between the aerosol layer and the filter matrix explicitly into account.
The principal 3-monthly check is an audit of the MAAP inlet flow. Record in check sheet and Particles Logbook. We have had over 80 young people take part, ranging in age from year 6 to year The winner of each category came from our core group! Personal tools Log in. Record details in the Particles log.
During normal operation, the keyboard usually is locked. Temperature and pressure calibration are described in section 4. Follow the checks in the last paragraph on page Here all about it here. Record initial and final readings in the Particles Logbook. We will be continuing the programme through to the end of this term and are hoping to continue bgl To calibrate the unit, a code must be entered to access calibration rights. Clean the grit-pot on the in-line cyclone and regrease lightly with silicone grease.
Observe the instrument display. Ignite was really excited to be invited to be a part of the sport component. The flow rate is then calculated from the pressure drop measured at the orifice plate together with pressure and temperaturebgl Figure 1.
BGI also bought trophies and spot prizes. Put MAAP into calibration mode will require the access code: Each Tuesday after school we have been running coaching, skills and drills and the opportunities for young people to play competitively. See below for some photos from the big. We chose to run a coaching programme around a sport called Spikeball that we recently started using. Scroll down to Barometric Pressure, record pressure indicated by MAAP and either station level pressure from Obs or from a traceable barometric sensor.
If required, replace the carbon vanes in the vacuum pump as per chapter 5 and 7 of the manual. Record bgk average reading in the particle log, the value should be Create a book Add wiki page Books help. Last week we gave them the opportunity to showcase the skills they have learnt by setting up a tournament. BGI works alongside various youth workers and tutors to provide opportunities for young people in areas such as: Be sure to check correct rotation of the spools and correct spool tension.
This core group have really progressed and developed into good Spikeball players. Exhaust port exhausted to outside laboratory, via bench-end extractor. Leave the filter in place for sufficient time to determine the zero level, this can be read by scrolling down the display to give last hour, 3 hour and 24 hour averages.
Hitting the Mark
Spikeball USA very kindly sponsored some sets and t-shirts to give away as prizes in the categories of best male and female high school and intermediate students.
Identifying and interpreting the error codes are described in Chapters 3 and 6 of the manual. With the keyboard enabled only operational changes are possible. Flow check as per Chapter 5 of the manual. Contents 1 Amendment Record Sheet 2 Introduction and description 2.
It uses a multi-angle absorption photometer to measure the radiation fields in the forward and back hemisphere of a glass-fibre filter caused by deposited particles. It is a skills based group-mentoring programme aimed at teenagers from the communities around the Walter Nash Centre in Taita.
Changing operational and calibration parameters are only possible after enabling the keyboard. Observe instrument output — check that data are within normal expected ranges.
Check if there are any instrument Status conditions or Errors present. Record all information in the Particles log. Flow rate in the MAAP is determined using an orifice plate located after the filter tape.